Ten (10) ReproEasy Pad gives you the ability to regularly verify the functionality of the MA37 or MA160 moisture analyzers. It guarantees reliable results during routine use. The ReproEasy Pad is a one(1) time use consumable item..
ReproEasy Pad Moisture Analyzer performance test pad only work with the Sartorius MA37 or MA160 moisture analyzer and recommended verses testing your moisture balance with sodium tartrate dihydrate
. One time use individual hydrogel pad for testing the performance of Sartorius MA37 or MA160 moisture analyzer. Contains acrylate taurate | PEG crosspolymer, glycerol, aqua. .
Enter the Calibration/Adjustment section of the menu on your Sartorius MA160. The settings for calibration and adjustment of the moisture analyzer can be defined here.
Select "MA performance test". The performance test can be used to check whether the analyzer has a fault or is ready for operation. During a performance test, both the heating unit and the weighing system are tested at the same time. You can run this test at either 70°C, 100°C or 130°C.
Open the packaging of the ReproEasy pad and remove the pad. Remove the protective wrapping and affix the pad to the center of the aluminum moisture analyzer sample pan
, sticky side down.
Close the hood on the moisture analyzer. The performance test starts automatically. During the performance test, the process status light on the analyzer flashes, if the light has not been switched off in the menu. The display shows the progress of the test and the remaining time.
If the performance test measurement is not successfully the display will show the message "Device is not OK". If you get this error we first recommend you calibrate the moisture analyzer with the required calibration weight (either 50 gram
for the MA37 or 100 grams
for the MA160 - the calibration weight
is not included with the moisture analyzer but instead an OPTIONAL Accessory). Then run a new performance test measurement is not successfully and if the message "Device is not OK" still appears contact Sartorius service.
If the test is successful, the message "Device is OK" first appears. Then the analyzer indicates that the sample can be removed and you are ready to perform moisture determination on your samples.
For additional information please refer to your Sartorius operational manual that came with your moisture analyzer.