microbalance Sartorius MSU3.6P-000-DM
Cubis polyrange microbalance (0 - 1.1g x 1ug) (1.1 - 2.1g x 2ug) (2.1 - 3.1g x 5 micrograms) with Black & White graphical display features modular Cubis design..
Availability: 0 left in stock. Usually ships in 2 months
YOU PAY: $20,064.00
Affordable and easy to use microbalance - Sartorius microbalance combining affordability with precision in the microgram weighing ranges. Sartorius innovative Cubis Series incorporating the highly advanced mechatronic monolithic weigh cell technology yields fast, repeatable, stable and accurate results.
Polyrange weighing offers greater readability for smaller sample weights. Here's how it works - In the high resolution mode and your weighing samples between 0 to 1.1 grams the balance will increment in 0.001mg intervals. When the capacity on the platform is in the range of 1.1 to 2.1g the display resolution will increment in 0.002mg intervals. Finally if your weighing samples 2.1 to 3.1 grams the display resolution will increment in 0.005 mg intervals. So as the weighing capacity increase the display resolution decreases. No matter how the balance is used the total cumulative weighing capacity on the weighing platform can never exceed 3.1 grams.
Highest level of reliability and standard conformity
Advanced Pharma Compliance makes Cubis ideal for use in regulated areas. Alert messages and reminder functions with user-definable action hierarchy for leveling, minimum sample weight, calibration and adjustment. Fully automatic calibration and adjustment with isoCAL, GLP-compliant, user configurable printouts.
Sartorius brings uncompromising quality in weighing technology to its new generation of Cubis microbalances. This micro balance combine fast response times with high repeatability when weighing the smallest of samples. Sartorius micros deliver results that meet the requirements of quality management systems everywhere and weigh with ultra-fast response times.
With a stabilization time of only 10 seconds, combined with resolution over the entire weighing ranges from 1 μg to 3.1 g Sartorius is the standard to be measured by in microbalances.
Microbalances by Sartorius offer "Intelligent" draft shield for easy weighing. With the "intelligent learning function", the motorized draft shield can be adapted to any work routine at the press of a key. It works when you want it to and not the other way around. A special conductive coating on the glass prevents the build-up of static electricity. All draft shield components can be removed from the balance for fast and easy cleaning.
The "isoCAL" function for fully automatic calibration and adjustment ensures consistently high accuracy and provides ISO/GLP-compliant user definable documentation and calibration record memory.
The SQmin function can be used to monitor minimum initial sample weight in accordance with USP standards.
The Cubis MSU balances offer a high resolution black and white 5.7" graphic display touch screen display with automatic round glass draft shield. This microbalances has 3 different levels of readability. The polyrange offers greater readability for smaller sample weights. Here's how it works - In the high resolution mode when you are weighing samples between 0 to 1.1 grams the balance will increment in 1 microgram graduations. When the capacity on the platform is in the range of 1.1 to 2.1 grams the display resolution will increment in 2 microgram intervals. Finally if your weighing samples 2.1 to 3.1 grams the display resolution will increment in 5 microgram graduations. So as the weighing capacity increase the display resolution decreases. The total cumulative weighing capacity on the weighing platform can never exceed 3.1 grams.
If you are a performing any manufacturing process that requires all weighments to comply with the mandatory USP Chapter 41 the MSA and MSU Cubis microbalances can be your solution to compliance. USP Chapter 41 titled "Balances" outlines "Repeatability" and "Accuracy", which specifies the operating range of a weighing device. The operating range is limited to the maximum capacity of the balance and begins at the point at which the balance's repeatability is less than or equal to 0.10%. The weighing must also be performed using a balance that is calibrated over its operating range.
Achieve maximum performance when used for inspection, measuring and test equipment within a quality system. The following features make your quality assurance procedures much easier: SQmin function: displays the allowable minimum sample weight according to the United States Pharmacopoeia (must be activated can calculated); Fully automatic calibration and adjustment feature, isoCAL; ISO/GLP-compliant recording; Input capability for alphanumeric sample IDs; and more.
The location of a microbalance is critical since environmental conditions can dramatically affect the weighing performance. Many people call us shopping for a microbalance and their only interest is what is the cost of the microbalance forgetting they might need a static eliminator and spend $1000's to change the location they planned on installing the microbalance. The higher the lab balance resolution or accuracy, the more care must be taken with environmental conditions.
Installation is always best on the ground level on a concrete foundation since installing the micro-balance on upper floor levels are more prone to the possibility of floor vibrations due structural motion and pedestrian movements. If you can't install your microbalance on the ground level installing the micro balance in a corner outside wall is the next best location. Always try to stay away from the middle of the floor since this area bends the most. Also the higher floor levels the balance is located, the greater the building sway and even slight misalignment along the vertical to the center of the earth can introduce substantial weighing error.
A high precision microbalance should always be installed in a low traffic area. Due to the expense of the micro balance it should be in a restricted area to lessen the chance of unauthorized people from touching or using the microbalance.
Installing a microbalance should be in quiet, temperature and humidity controlled, draft-free location is of paramount importance. Finding a location in cleanroom can many times be challenging due to laminar air flow. So before even considering purchasing a high precision laboratory balance you should consider the area you want to install the microbalance since if your environment is not ideal you don't have a defective microbalance but a defective environment. You can read more about selecting a location of your lab balance here.
A solid balance table or marble slab is recommended to eliminate the surface the microbalance is installed on to prevent bending and any flexing. The weighing table should also be decoupled from the environment with an additional layer of shock-absorbing material such as rubber or cork to lessen sensitive to shock and vibrations. Wooden tables have some degree of flexure and typically unsuitable to attain repeatable consistent weighing results with a microbalance. Marble tables should not only have a marble top but also marble legs and these tables typically weigh several hundred pounds and cost well over $1000 and installation and freight will also be costly. Please consider these issues are additional costs and many times required for the microbalance to perform to the capabilities of the precision weighing instrument.
Controlling temperature changes is the most critical environmental condition that you must control and has the most serious affect on the performance of your microbalance. Temperature effects of even a difference of 1°C can cause sensitivity and zero point to shift. Although some people might say zero point drift can be resolved by taring the microbalance sensitivity drift will introduce systematic error. Sensitivity drift for the Sartorius MSU3.6P000DM is stated at ± 1 ppm / K. " K " is parts per million per °C rise in temperature. So if the temperature changes by 1°C you can expect 1 microgram change. Since the MSU3.6P has readability to 1µg just a 1°C temperature change will affect the microbalance by one (1) divisions or 1µg. A temperature change of 2°C might not sound like a lot but if you placed a sample of 0.000010 grams (ten micro grams) a difference of 2°C will affect the observed weight of 0.000010 gram mass by up to 2 micrograms. So instead the the microbalance reading 0.000010 grams it could display between 0.000012 grams to 0.000008 grams. That is a lot of error just over a temperature change of 2°C and the lighter the sample the more significant the error.
Once you have an adequate environment installing the microbalance isn't difficult, but for those customers wishing for professional installation it is available at an additional cost. Please call us at 978-521-7095 for current prices for these services, but the average cost is $425 in the lower 48 states of the USA plus travel and expenses. IQ/OQ testing is also available for an average cost of $875 plus travel and expenses.
Low humidity conditions tend to promote static electricity and causes havocs with your weight readings. Air conditioning removes moisture from air and can be one of the culprits to your static issues. Weighing samples in plastic, glass or Teflon containers can also contribute to static issues. We offer a large selection of disposable microbalance weigh dishes and always recommend the use of these aluminium pans to lessen the chance static can affect your weighing results.
The Sartorius MSU3.6P000DM microbalance is made to order in Germany with a typical lead time of 6-8 weeks. To check availability give us a call for current delivery times. If this lead time is unacceptable we would recommend you consider the preconfigured MSA3.6P0TRDM which Sartorius tries to keep in stock.
Purchasing a micro balance is an extremely costly purchase and sometime even more expensive is creating the proper environment. If you have any technical questions or want to check availability give the balance experts at Precision Weighing Balances a call at 978-521-7095.
Free shipping in the lower 48 states for the MSU3.6P000DM micro balance.
|Capacity x Readability
(grams x µg)
|0-1.1 x 1µg
1.1-2.1 x 2 µg
2.1-3.1 g x 5µg
|Repeatability, micrograms||3 / 4 / 5 µg|
|Linearity, micrograms||±4 µg|
|Corner Load (mg)*
(Test Load [g])
|± 5 mg (1 g)|
|Minimum Sample Weight (mg)||Typically 4 mg minimal initial weighing according to USP (United States Pharmacopoeia), USP31-NF26|
|Tare Range||FULL to Capacity|
|Stabilization Time (avg)||< 5 seconds|
|Response Time (avg)||< 8 seconds|
|± 1 ppm / K|
|Selectable Weight Units||g, kg, ct, lb, oz, ozt, tlh, tls, tlt, GN, dwt, mg, /lb, tlc, mom, K, tol, bat, MS and parts per pound|
|Display Update||0.1 – 0.4 seconds
(Depends on The Filter Level Selected)
|Display||High Resolution black and white 5.7" graphic display|
|Adjustment of the display and control unit||Display tilts; Control unit and display detachable|
|8 Selectable Application Programs||Unit conversion, isoCAL automatic calibration/adjustment function, density determination (buoyancy method only), calculations, averaging, net/total formulation, weighing in percent, counting|
|10 Additional Selectable Application Programs||SQmin function for minimum initial weight according to USP, individual identifiers, statistics, time-controlled functions, totalizing, DKD measurement uncertainty, second tare memory, checkweighing, alibi memory, audit trail|
|Allowable Ambient Operating Temperature||5°C to 40°C
(41°F to 104°F),
80 % for temperatures up to 31°C, decreasing linearly to 50 % relative humidity for 40°C
|Motorized Internal Calibration||Yes|
|External Calibration||3 g Class 1 (not included)|
|Weighing Pan Size||ø 30 mm / 0.78 inches|
|Draft Shield Height||70 mm / 1.2 inches|
|Protection||Protected against dust and water|
|Standard interface ports||USB (built into weighing module), RS-232C port for connecting accessories - 25-pin (built into weighing module)|
|Ethernet port||Yes - built into display and control unit|
|SD Card Reader||Yes - Built into display and control unit|
|Control Unit Dimensions WxDxH||9.4"
x 10.9" x 2.2"
240 x 277 x 56 mm
|Weigh Unit Dimensions WxDxH||5.2"
x 13.4" x 4.7"
132 x 340 x 121 mm
|Power Source||AC Adapter 120 VAC included with scale|
|Standard Equipment||120V AC Adapter, level indicator, adjustable feet, RS-232 and USB interface, hanger for weigh below, manual|
|Optional Accessories||Data printer for connection to RS-232, 25-pin RS232C data interface cable, SartoCollect software for data transfer, Cal-Pak 3 piece calibration weight set 2g, 200mg, 1mg|
|Warranty||2 years manufacturer's warranty|
YOU PAY: $119.91
YOU PAY: $4,073.61
YOU PAY: $26,671.25
YOU PAY: $10,719.00
YOU PAY: $4,546.39
YOU PAY: $33,877.00
YOU PAY: $4,588.00
YOU PAY: $26,851.75
YOU PAY: $31,929.50
YOU PAY: $3,753.56
YOU PAY: $11,650.50
YOU PAY: $34,570.50
YOU PAY: $33,696.50
YOU PAY: $118.65
YOU PAY: $25,977.75
YOU PAY: $137.44
YOU PAY: $4,771.61
YOU PAY: $32,803.50
YOU PAY: $12,217.50
YOU PAY: $20,757.50
YOU PAY: $27,450.25
YOU PAY: $16,914.55
YOU PAY: $9,135.00
YOU PAY: $25,778.25
YOU PAY: $33,003.00
YOU PAY: $129.99
YOU PAY: $10,633.50
YOU PAY: $24,904.25
|Capacity x Readability||0-1.1 g x.1 ug|
|Capacity x Readability (dual range)||2.1-3.1g x 5 ug|
|Readability 2||0.000005 g|
|Weight Modes||Carats, Grains, Milligrams, Pounds, Ounces, Pennyweight, Grams, Troy Ounces, KiloGrams|
|Functions||Averaging, Formulation, Percent Weighing|
|Features||Parts Counting, RS-232c, Internal Calibration, USB, Ethernet|
|Legal for Trade||No Legal for Trade|
|Construction||Solid Diecast Metal Base Housing, Stainless Steel Weighing Pan, LCD Display, Magnetic Force Motor Monolithic Weigh Cell|
|Calibration||Motorized automatic internal calibration|
|Communication||RS-232, USB, Ethernet|
|Platform size (inches)||0.8 diameter|
|Power Requirements||120V AC Adapter included|
|Manufacturer Warranty (years)||2|
|Additional info||Cubis polyrange microbalance (0 - 1.1g x 1ug) (1.1 - 2.1g x 2ug) (2.1 - 3.1g x 5 micrograms) with Black & White graphical display features modular Cubis design..|